Technical Spares. User Manual, GSI Product Code: Other Product Code: Manufacturer: Grason Stadler. Instruction Manual, GSI Direct calibration for TDH headset and bone vibrator. THE COMPLETE CLINICAL AUDIOMETER. AND MORE. The GSI 61 comes with everything you’ll need to. 2 GSI 61 NEW STANDARD IN CLINICAL AUDIOMETRY The GSI 61 is the versatile, making hearing aid evaluation easier than ever Direct calibration for TDH.
|Country:||United Arab Emirates|
|Published (Last):||28 May 2011|
|PDF File Size:||14.28 Mb|
|ePub File Size:||8.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
When the Interrupt is in the off position, the corresponding channel is activated by depressing the Tone bar bar and deactivated by releasing the Tone bar.
This GSI 61 Includes: Screening Procedure Assist the patient with the placement of earphones.
Philips HD5 ultrasound system Reliable versatility Philips HD5 ultrasound system Affordability and features in one Designed to perform as you need it Every day, your patients come to you for high-quality care.
Bone Vibrator Check This check must be performed in a quiet environment or in a sound room. Purpose The purpose More information. As a pioneer of More information. These pushbuttons are also used to select the intensity increase when testing in the SISI mode.
Select the High Frequency audiogram display format by pressing the High Freq pushbutton on the front panel.
Grason-Stadler (GSI) – Audiology Solutions Network
Times up to minutes and 59 seconds may be displayed before the timer resets to 0: The hearing levels listed in this table are guaranteed maximum levels.
Set the Channel 1 attenuator to dB. The following is the list of status messages which will occur with operation of the GSI 61 Audiometer. The High Frequency earphones are connected to stereo cable extensions which, in turn, are connected into the Right and Left phone jacks on the rear panel. Present one word at each decrement.
Using either live voice or recorded speech, present the standardized word lists W-1 and W-2 Appendix 1 in the guidetesting the better ear first. Reliable versatility Philips HD5 ultrasound system Affordability and features in one Designed to perform as you need it Every day, your patients come to you for high-quality care.
No part of this publication may be reproduced or transmitted in any form or by any means without the prior written permission of Grason-Stadler Inc. Unpacking Carefully remove the GSI 61 from its shipping container. It s your vital connection to current customers and new.
Input Record Type 5 – Pushbutton Code Record This record type provides the ability to remotely simulate the operation of all user controls. The display indicates the number of correct responses and the total number of presentations along with the percentage of those correctly identified. Frequencies between and Hz are suggested as the most useful for the purposes of gei test.
Phones High Frequency Option Setup: Retest Hz followed by Hz and Hz.
The leader in diagnostic audiometry
The signal is manually alternated from frequency to frequency. Actual data capture and transfer occurs when the Data Transfer button on the front panel is pressed. If the instrument appears to have suffered mechanical damage, notify the carrier immediately so that a proper claim can be made.
If the bone vibrator cannot manuak calibrated to specifications, it should be replaced.
Grason-Stadler GSI 61 Manual Rev B
When Speaker is selected on both channels and the optional booster amplifier is not installed, it is possible to increase Channel 2 to a higher hearing level if Channel 1 is lowered. Phonak wireless add-ons Boost the performance of your hearing aids Phonak wireless add-ons Modern hearing aid technology makes hearing and understanding in most situations enjoyable. If there is currently no saved data, this error message will be flashed: These pushbuttons allow the operator to change the intensity step size to 1.
Introduction from 8 kHz to 20 kHz for ototoxic drug monitoring. Product Choose a Product A. The screening test typically takes minutes to complete. Seat the patient in a sound field room 6 feet by 6 feet at a distance equal to one meter from the speakers. Print Set configuration switch S according to the desired print color and audiogram arrangement.
Used GRASON STADLER GSI Audiometer For Sale – DOTmed Listing #
This record acknowledgment will not be sent if the GSI16 compatible record format has been selected.