The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||3 March 2010|
|PDF File Size:||2.23 Mb|
|ePub File Size:||7.81 Mb|
|Price:||Free* [*Free Regsitration Required]|
Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. Preparing For A Test 1.
Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1. Wurster M, Doran T. Keep the tube level, and allow it to fill by capillary action. By pressing ‘print’ button you will print only current page.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
You may see the following error messages while using the The meter is too cold or too Turn the meter cs and allow it CoaguChek XS Meter. To exit Erase mode without Coguchek erase all stored results: Erasing Stored Test Results 2.
Page of Go. A rotation scheme was used to ensure that the same instrument was not always dosed first. The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe.
Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. Record Result Record the result according to test strip.
Table Of Contents User Manual Identify the Target Area Find the target area on the 7. Code Chip Code Chip 2.
Using Venous Blood 2. Performing A Test 3. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Don’t show me this message again.
Fill the Capillary Tube When the flashing test strip 4. Page 24 For more information, see re-insert it.
Enter text from picture: If the memory is result in memory. Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. The CoaguChek XS PT test strip may be used for patients under a combination therapy of oral anticoagulants coagucheo heparin injections.
View Most Recent Result The most recent test result appears. For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
Dose the Test Amnual Discard the first 4 drops of blood. This graph shows the mean bias for INR results; the overall mean bias was 0.
Roche CoaguChek XS User Manual
If the numbers are different, Close the container tightly. Got it, continue to print. Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip.